On occasion, you may notice a 'step' appearing in your CO2 graph within the Tether portal.
This is the CO2 sensor recalibrating itself.
This will usually happen within the first two weeks of receiving a Tether device. Before a device is shipped to you it sits in a warehouse unpowered. This can cause the calibration of the CO2 sensor to drift. During the first two weeks of operation the CO2 sensor will constantly search for "fresh air" - which is around 400ppm. After two weeks the CO2 sensor will recalibrate itself and this can often result in a drastic sudden spike in a CO2 reading.
To force the Tether sensor to calibrate itself, take the sensor outside and press and hold the reset button for 5 seconds until a solid light appears. This will tell the sensor that the air is fresh/outside air (400ppm) and it will calibrate itself to that reading.
It is recommended to perform this action before you install a Tether sensor.
The readings taken before the first calibration will usually be high and not accurate. This is why it is recommended to reset the Tether device outside before installing.
Please sign in to leave a comment.